Vibration measuring device VIB-8; echo "";
The devices have 8 measuring channels with successive internal commutation. The devices additionally provide stabilized supply current, necessary for the operation of piezoelectric sensors with an electronic preamplifier (Integrated Circuit Piezoelectric — ICP).
Commutation of the measuring channels as well as AD conversion and processing of the measuring results is carried out for each channel successively. Commutation of the measuring channels, digital processing, storage and exchange of information via RS-485 (Modbus RTU) and Ethernet (Modbus TCP/IP) interfaces is carried out under the control of microprocessor.
The devices are applied for control and balancing of the mechanisms in oil, gas, electrotechnical and other industries which use rotary units (gas-, steam- and hydro turbines, compressors, pumps, electric motors etc.).
|1||AC voltage rms value measuring range within 5-3000 Hz frequency measuring range (if there is a constant component of electrical voltage within the range of 8-12 V)||V||from 0,0007 to 5|
|2||Reduced measuring error allowable limits of AC voltage rms value||%||±1,0|
|3||DC power value for power supply of piezoelectric sensors||mА||11±3|
|4||Coefficient of vibration acceleration scale transduction||m/(Vs 2)||100|
Vibration parameters measuring range, if primary transducers are used:
- vibration acceleration
- vibration speed
from 0,1 to 500
from 0,01 to 20
|6||Supply voltage from DC source||V||24±2,4|
|7||Power consumption, max||W||4,0|
Dimensions (length х width х height), max
- ambient temperature
- atmosphere relative humidity (temp. 30°С)
от -40 до +60
|11||Mean time between failures, max||h||40 000|
|12||Average lifetime, max||years||15|